New to MyHealth?
Manage Your Care From Anywhere.
Access your health information from any device with MyHealth. You can message your clinic, view lab results, schedule an appointment, and pay your bill.
The minimal scan duration needed to obtain reliable lesion volumes with computed tomographic perfusion (CTP) has not been well established in the literature.We retrospectively assessed the impact of gradual truncation of the scan duration on acute ischemic lesion volume measurements. For each scan, we identified its optimal scan time, defined as the shortest scan duration that yields measurements of the ischemic lesion volumes similar to those obtained with longer scanning, and the relative height of the fitted venous output function at its optimal scan time.We analyzed 70 computed tomographic perfusion scans of acute stroke patients. An optimal scan time could not be determined in 11 scans (16%). For the other 59 scans, the median optimal scan time was 32.7 seconds (90th percentile 52.6 seconds; 100th percentile 68.9 seconds), and the median relative height of the fitted venous output function at the optimal scan times was 0.39 (90th percentile 0.02; 100th percentile 0.00). On the basis of a linear model, the optimal scan time was T0 plus 1.6 times the width of the venous output function (P<0.001; R(2)=0.49).This study shows how the optimal duration of a computed tomographic perfusion scan relates to the arrival time and width of the contrast bolus. This knowledge can be used to optimize computed tomographic perfusion scan protocols and to determine whether a scan is of sufficient duration. Provided a baseline (T0) of 10 seconds, a total scan duration of 60 to 70 seconds, which includes the entire downslope of the venous output function in most patients, is recommended.
View details for DOI 10.1161/STROKEAHA.116.014177
View details for Web of Science ID 000389424200027
View details for PubMedID 27895299
View details for PubMedCentralID PMC5134896